Sartopore® 150 5441358K4--OO--BFiltration Area: 150 cm²
Ready-to-use disposable capsules for sterile filtration and reliable Mycoplasma removal in laboratory scale or for small scale filterability trials.
- Smallest Pleated Filter Element
- Ideal for Scale-Up & Scale-Down
- Proven Performance of Sartopore® 2
Sartopore 2 150 is a disposable, pre-sterilized, ready-to-use membrane filter capsule for highest convenience. Sartopore 2 150 capsules are made of a unique hydrophilic Polyethersulfone membrane providing outstanding total throughput|flow rate performance, low extractables and broadest chemical compatibility. The polyethersulfone membrane is compatible with a pH range from pH 1 to pH 14 making Sartopore 2 150 ideal for filtration of solutions with high|low pH. The unique pleated filter construction combined with the highly asymmetric pore structure of the Polyethersulfone membrane offers excellent flow rates and superior total throughput performance, especially in comparison to conventional stacked disc filter systems. Sartopore 2 150 capsules are available with two different pore size combinations and various connectors for simple installation in your filtration system. A Tri-Clamp connection assures secure and reliable integrity testing. Sartopore 2 150 capsules are designed, developed and produced under a DIN/ISO 9001 certified Quality-Management-System. Each unit is tested for integrity product quality and security. A Validation Guide is available.
Typical applications include sterilizing grade filtration and Mycoplasma removal from:
- Animal Sera
- Cell Culture Media
- Media Components
- Bioprocessed Pharmaceuticals
- Biological Fluids
- Any other application requiring sub 0.2 μm filtration for enhanced sterility assurance
The polyethersulfone membrane is compatible with a pH range from pH 1 to pH 14 making Sartopore® 2 150 ideal for filtration of solutions with high | low pH.
The unique pleated filter construction combined with the highly asymmetric pore structure of the polyethersulfone membrane offers excellent flow rates and superior total throughput performance, especially in comparison to conventional stacked disc filter systems.
Easy to Use:
Sartopore® 2 150 capsules are available with hose barb, 1/4 inch NPT-thread or 1 1/2 inch Tri-Clamp connectors for simple installation in your filtration system. The Tri-Clamp connection assures secure and reliable integrity testing.
The vent design enables easy access to the venting valve. A hydrophobic PTFE membrane positioned on the highest point upstream allows an easy venting of the capsule and prevents product loss during the venting process.
Featuring the same materials and type of construction as any other Sartopore® 2 filter element, Sartopore® 2 150 is ideally suited for R&D Labs in pharmaceutical development. Filtration trials can be performed using extremely small volumes of high value products.
Sartopore® 2 150 capsules are fully validated as sterilizing grade filters according to HIMA and ASTM F-838-05 guidelines.
Each individual element is integrity tested by diffusion and bubble point test prior to release, assuring absolute reliability.
Sartopore® 2 150 capsules are designed, developed and manufactured in accordance with a ISO 9001 certified Quality Management System. A Validation Guide and Extractables Guide is available for compliance with regulatory requirements.
Pharma / Biotech
|Membrane filter material||
|Max. Differential Pressure||
20 °C: 4 bar; 80 °C: 2 bar
Quantitative retention of 1 x E7 CFU/cm² Brevundimonas diminuta ATCC® 19146 per ASTM methodology
<0.18 EU/mL as determined by the LAL test
Product meets or exceeds the Quality standards set for "Sterile Water for Injection" by the current USP.
All materials of this filter element meet the requirements of the current USP Biological Reactivity tests <88> for plastics Class VI, (Systemic Injection, Intracutaneous and Implantation tests).
Autoclaving: Min. 25 cycles at max. 134 °C, 2 bar | 29 psi, 30 min.
This product complies with the title 21 of the Code of Federal Regulations (CFR), section 210.3(b)(6) and 211.72
The product was manufactured in conformance with established Current Good Manufacturing Practice (cGMP) standards.
|Max. allowed air diffusion||
1 ml/min at 4.0 bar | 58 psi at 20°C
Hydrophilic, heterogeneous double layer polyethersulfone membrane
1/4'' Hose Barb
1/4'' Hose Barb